The quality of closeness to a specified value under stated conditions. Usually expressed as uncertainty - the maximum deviation to be expected
between a reading or output and the actual value being measured.
A/D (ANALOG TO DIGITAL) CONVERTER
A circuit or device that produces a digital output representing the magnitude of an analog input signal.
A construction utilizing an air vane to achieve movement damping. This vane is usually housed in a closed chamber to increase the damping action.
Average temperature coefficient of RTD resistance over the interval of 0 to 100°C. Usually 0.00385?/?/°C for platinum.
The average temperature of air surrounding the device or equipment.
A parameter which varies in a continuous, rather than incremental or discrete-step manner.
A scale and pointer meter capable of indicating a continuous range of values.
ANSI C39.1 STANDARD
American National Standards Institute specification for analog meters.
A device that reduces the amplitude or power level of a signal without introducing appreciable distortion.
The ability to measure DC values of either polarity without the need to interchange test lead connections.
An automatic correction for offsets and drifts at zero input.
An AC measurement obtained using a DC instrument with a rectifying input circuit calibrated in terms of the corresponding RMS value. Accurate only
for pure sine wave inputs.
The change in the position of an analog pointer from zero when the axis of the moving element moves from the vertical position to the horizontal
position. The balance is expressed as a percentage of the scale length.
BANDWIDTH (FREQUENCY RESPONSE)
The frequency span where a constant amplitude input will produce a meter reading within a specified limit (usually 3db). In controllers, the region
around the setpoint where control occurs.
Current that flows out of an amplifier's input terminals which will produce a voltage drop across the source impedance. In a perfect amplifier this
error term would be zero.
Digital transmission speed in bits per second.
The input impedance of a measuring circuit (expressed in ohms) or the load on the secondary of a transformer (expressed in volt-amps or watts). In
potential or current transformers, burden is the maximum load the transformer can support while operating within its accuracy rating.
To determine the indication or output of a device with respect to a standard.
Temperature scale where 0° = freezing and 100° = boiling point of water at sea level. Formerly known as Centigrade.
CMR (COMMON-MODE REJECTION)
The ability of a circuit or meter to reject a signal that appears at both input terminals with respect to ground.
CMV (COMMON-MODE VOLTAGE)
An AC or DC voltage which appears between the signal lines and circuit ground or earth.
COLD JUNCTION COMPENSATION
A correction applied to thermocouple measurements to compensate for the temperature of the TC wire connections, so the temperature reading is
only the result of the measuring TC junction.
The ability of a substance to transmit heat by conduction.
The difference between the actual response and the ideal response to a particular stimulus.
A special polarized disconnect device whose current-carrying parts are of thermocouple alloy material.
A device capable of receiving a signal from a process and regulating an input to that process in order to maintain a selected operating condition
Type of control used in a feedback control system. One mode is proportional control. Two mode is proportional plus integral (reset) or derivative
(rate). Three mode is proportional, integral and derivative (PID).
The number of analog-to-digital conversions performed per second by a digital instrument.
One event or one increment of the least significant digit.
Central Processing Unit in digital computing systems. Often referred to as microcontroller or microprocessor.
The ratio of the maximum (crest) value of a periodic function (AC voltage or current) to its RMS value.
The manner in which the pointer of an analog instrument settles at its steady indication after the applied electrical energy is changed. Usually
expressed as percent over-shoot.
A meter movement based upon the permanent magnet DC motor principle: a small coil of wire supported on jewel bearings or taut band between the
poles of a permanent magnet. The magnetic field of the DC current passing through the coil interacts with the magnet's field, causing rotation of the
coil and an attached pointer against the restoring force of coil springs.
The standard unit for measuring relative strength of a signal parameter such as power or voltage. dB= 10 log (Pout/Pin); dB=20 log (Vout/Vin). 20dB
corresponds to a voltage ratio of 10. A NMR of 70dB provides voltage noise rejection of 3162/1.
The region through which an input can be varied without initiating a response.
The voltage that can be sustained without breakdown.
An input circuit where signal high and signal low are electrically floating with respect to signal common or signal ground.
A measure of the display span of a meter. By convention, a full digit can assume any value from 0 through 9, a 1/2 digit will display a 0 or 1 and
overrange at 2, a 3/4 digit will display up to 3 and overrange at 4. A meter with a display span of ±3999 counts is a 33/4 digit meter.
DIN (DEUTSCHE INDUSTRIE NORM)
A set of German technical standards. Commonly used to specify panel meter sizes.
An unwanted change in the reading or setpoint value over time, when inputs are held constant.
An analog to digital conversion technique which can provide high noise rejection.
Concurrent transmission and reception. Contrasts with half-duplex (one direction at a time).
ELECTRO MOTIVE FORCE (emf)
An electrical potential difference which produces or tends to produce an electric current.
A circuit in a watt, VAR or PF meter that accepts one voltage and one current input.
An arrangement that expands a specific portion of an overall range to occupy a larger portion of the full-scale length than it normally would.
An enclosure capable of withstanding an explosion of a specified gas or vapor which may occur within it and of preventing the ignition of a specified
gas or vapor surrounding the enclosures by sparks, flashes, heat or explosion of the gas vapor within.
Assuming a safe operating mode in the event of a failure.
A method for specifying relay contact configuration: Form A SPST-NO single pole, single throw, normally open Form B SPST-NC single pole, single
throw, normally closed Form C SPDT single pole, double throw Change-Over CO same as SPDT
FULL SCALE VALUE
The arithmetic sum of the two end-scale values (may not apply to some specialized meters, such as power factor). When zero is not on the scale,
the full-scale value is the higher end-scale value. Examples: Scale Full Scale Value 0-200V 200V 25-0-175V 200V 250-0-250V 500V 60-160V 160V
A continuously changing temperature as a function of distance.
Reference point for an electrical system. Often used to indicate an earth connection or negative side of a DC supply.
A thermocouple construction where the junction is attached (grounded) to the sheath as contrasted to an ungrounded or exposed junction type.
The difference in an output or activation point due to rising vs. falling input signals.
The total opposition to electrical flow, including both resistive and reactive elements.
INPUT RESISTANCE (INPUT IMPEDANCE)
DC (or AC) resistance measured across the input terminals with signal leads disconnected.
The resistance measured between two insulated points on a device when a specified DC voltage is applied.
International Temperature Scale, Revised 1990. The currently accepted thermodynamic temperature scale. Replace the previous standard IPTS-68.
A technique for measuring AC or DC current using mutual repulsion of magnetic fields. The rotation is proportional to the strength of the fields, which
is proportional to the current flowing in the field coil. Also described as 'moving vane', 'moving iron' or 'repulsion'.
A process or area that is a constant temperature.
The point in a thermocouple where the two dissimilar metals are joined.
The basic temperature unit of the thermodynamic scale. 0°C= 273K
Analog meter pointer with end flattened and turned edgewise so that the thinnest dimension or edge is seen by the observer. Often used with
mirror-backed scale for increased reading accuracy by elimination of parallax.
LATCHING (in meter relays)
A condition that requires the manual reset of a tripped relay. The tripped relay cannot be reset (re-energized) until the indicating pointer or display is
in a non-alarm position.
Insensitivity to a power line frequency interference signal. Usually expressed in dB.
Technique for minimizing or eliminating errors due to signal leads.
A measure of the departure from a straight-line response in the relationship between two quantities, where the change in one quantity is directly
proportional to a change in the other quantity. Usually expressed as a maximum percent.
LSD (LEAST-SIGNIFICANT DIGIT)
The rightmost active digit of a digital display.
That junction of a thermocouple subjected to the temperature to be measured.
An analog meter scale with a mirror arc that enables alignment of the eye's line of sight perpendicular to the scale when taking a reading. Eliminates
parallax, considerably improves reading accuracy.
MOV (Metal Oxide Varistor)
Component used to protect electronic circuitry from voltage transients.
MSD (MOST SIGNIFICANT DIGIT)
The leftmost digit of a digital display.
A technique which allows different input (or output) signals to use the same lines at different times, controlled by an external signal. Multiplexing is
used to save on wiring and I/O ports.
A rating standard from the National Electrical Manufacturers Association which defines intended use and degree of environmental protection.
A US case standard for digital panel meters, which requires a panel cutout of 3.93 x 1.69 inches.
NMR (NORMAL-MODE REJECTION)
The ability of a meter to filter out noise that appears across the signal inputs in the same manner as the desired signal. Also known as series mode
In an ideal system, the input-out relationship is linear (i.e., straight line). Any departure from straight line is expressed as non-linearity. Two methods
are used for measurement. The "best straight line" approach compromises the end points and situates the line to give the most optimistic answer.
The non-zero output of a device for zero input.
The difference between two bias currents drawn by the inputs of a differential amplifier.
OHMS PER VOLT
Indication of the total terminal resistance of an analog voltmeter. A 1000-ohms-per-volt meter has a resistance of 150,000 ohms on its 150-volt
(full-scale) range, and 300,000 ohms on its 300-volt range. Its basic movement is a 1mA meter.
Non-proportional control in which the controlled process input is either fully on or fully OFF depending on whether the temperature is above or below
the control point dead-band.
An isolation device that provides an electrical barrier between related circuits.
The excess load beyond full-scale value that an instrument can withstand without damage or failure. Expressed as a percent of a full-scale value.
A reading that exceeds full scale. OVERSHOOT The amount by which a meter or process exceeds the final value during a transition. Usually
expressed as percent of amplitude for a step change.
An instrument function that permits retaining and displaying the peak value momentarily reached by a signal.
The difference in electrical degrees by which current leads voltage in an inductive circuit or lags voltage in a capacitive circuit. Also the phase
displacement between primary and secondary currents in a current transformer.
PIVOT AND JEWEL
Method of suspending the moving coil or moving iron vane in a magnetic field, in analog meters. The movable element is equipped with two metal
pivots which engage glass or synthetic-jewel cup bearings.
A wattmeter with 2 or 3 single-phase wattmeters mounted in the same package.
The relative voltage at a point in a circuit or in space with respect to some reference point.
The ratio of consumed power to apparent (volt-ampere) power in an AC circuit.
PPM (PARTS PER MILLION)
A convenient format to express very small numbers, such as temperature coefficients. 100ppm is 0.01%.
A panel meter with sizeable zero and span adjustment capabilities, which can be scaled for readout in engineering units for signals such as 4-20mA,
10-50mA and 1-5V.
PSIA (PSI Absolute)
A pressure reading using vacuum as the reference.
PSIG (PSI Gauge)
A pressure reading using ambient air pressure as the reference.
The span of values over which a meter will function without entering overload condition, e.g. 0-150VAC, 0-10A.
A resistance measurement technique where the unknown resistance is placed in series with a known resistance. The voltage across each is
measured to determine the unknown resistance.
The opposition presented by capacitance and/or inductance to the passage of alternating current of a given frequency.
A DC meter equipped with a solid-state rectifier at its input to convert AC energy to DC energy. The instrument provides measurements of the
average value of an AC voltage or current, and its scale is usually calibrated in terms of the RMS equivalent. Such calibration is accurate for pure
sine-wave signals, but the accuracy decreases for distorted signals.
The other junction (usually at ice point) to which the measuring thermocouple junction is compared (see cold junction).
The ability of an instrument to register the same reading in successive measurements of the same input.
The degree to which nearly equal values of a quantity can be discriminated. In digital meters, the value represented by a one-digit change in the
RMS (ROOT MEAN SQUARE)
The square root of the mean of the square of the signal over one full cycle. Effective heating value.
RTD (RESISTANCE TEMPERATURE DETECTOR)
A metallic sensor where resistance increases in a predictable manner with increasing temperature.
An unwanted connection between a pair of thermocouple wires tending to produce a signal representative of the secondary junction temperature
rather than the measuring junction temperature.
Internal heating of a transducer as a result of power dissipation.
The minimum change in input to which a device can respond.
The value of a process variable that will initiate an action. The desired control point.
The time required for the output to settle within a specified band of the final value when a step input change is applied.
A protective conductive covering that provides a least resistance path to ground for external interference.
A calibrated low resistance connected in parallel with the input terminals of an ammeter in order to enable measurement of higher currents. It can be
internal or external. Typical external shunts are either 50mV or 100mV full scale.
S.I. (SYSTEM INTERNATIONALE)
The formal name for the metric system.
A circuit or module which offsets, attenuates, amplifies, linearizes and/or filters the signal for transmission or processing by an A/D converter. The
typical output span of a signal conditioner is ±2VDC or 4-20mA.
SINGLE ENDED INPUT
Amplifier with one input referenced to circuit common or ground.
A resistance/capacitor or diode/resistor network used to dissipate switching transients. Often used across high current relay contacts.
The ability to adjust the gain of a process or meter so the display span corresponds to a specified signal span.
An AC motor whose speed is exactly proportional to the frequency of the applied alternating voltage.
Method of suspending moving coil or moving iron vane in magnetic field. Eliminates pivot and jewel friction problems.
TEMPERATURE COEFFICIENT (TEMPCO)
The change in a parameter produced by a change in temperature. Normally expressed in %/°C or ppm/°C.
A semiconductor material which exhibits a known electrical resistance vs. temperature.
The junction of two dissimilar metals with a voltage output proportional to temperature.
THERMOCOUPLE BREAK PROTECTION
A means to indicate when thermocouple has failed in an open circuit condition.
THERMOCOUPLE LOOP RESISTANCE
The total resistance of the thermocouple and its extension wire.
A number of thermocouples connected in series, arranged so that alternate junctions are the referenced temperature and at the measured
temperature to increase the output for a give temperature difference between the measuring and reference junctions.
The housing into which an RTD or thermocouple is inserted. Allows easy removal and/or replacement.
The time required for a sensor to respond to 63% of its total change resulting from a step input. Five time constants are required to attain 99% of
the total change.
A logic output which has 0, 1 and high impedance output states. For parallel connected outputs, the high-impedance state is used when the output
is not active.
TRUE RMS (TRMS)
The true root-mean-square value of an AC or AC-plus-DC signal. Often used to determine power of a signal.
A signal conditioner in which the signal output and power input share two wires, thus minimizing wiring.
In electronic meters, a capability for measuring voltage or current of only one polarity, [i.e., the (+) or (-)].
VAR (VOLT-AMPERES REACTIVE)
The unit of reactive power, as contrasted to real power (watts).
The unit of electromotive force. One volt applied to a resistance of one ohm produces a current of one ampere.
Unit of real (effective) electrical power. W = VA x PF in a sinusoidal circuit.
The ability to adjust a signal conditioner or meter so that zero output or zero display corresponds to a specific input signal, such as 0V or 4mA.